F.-P.Schilling BDC Status

BDC Status

Data Quality Meeting, 26/10/99 Frank-Peter Schilling

HV Status:

• Since 21/10/99, Fill 2146, Run 257364 outer half-sector in layer 4, sector 3 is dead and disconnected from HV

• Now 5 out of 128 halfsectors dead (→ 3.9%) • Still at least 7 layers available in full θ, ϕ-region!

F.-P.Schilling BDC Status

LV Status / Noise:

• Noise in Layers 4/5 due to faulty LV Power Supply L4 Histogram: DIGI-NO-ANA

• PS (−5.0 V ) exchanged on Saturday evening • Noise occasionally with replacement PS • Original being repaired now, put back in soon